J Neurosurg Pediatrics 5:4–16, 2010.(DOI: 10.3171/2008.7.17690)
This 25-year follow-up study was performed on 120 children with hypertensive hydrocephalus to evalu- ate the influence of the early prophylactic implantation of the Integra antisiphon device (ASD, Integra Neurosciences Ltd.) on the rate of proximal shunt obstructions and the frequency of symptomatic slit ventricle syndrome (SVS). The adaptability of the ASD to growth, proper positioning of the ASD as a necessity for its successful performance, and the 3 phases of SVS development are discussed.
Method. Since 1978, the ASD has consistently been implanted either at the time of primary shunt insertion (66 neonates, mean follow-up 11 years) or during revisions of preexisting shunts (54 children, mean follow-up 11.8 years). The complication rate among the 54 children before ASD implantation (mean follow-up 8.3 years) was com- pared with that among all 120 patients once an ASD had been inserted. Shunt complications were documented as ventricular catheter, distal catheter, and infectious complications.
Results. The study revealed a significant long-term reduction in ventricular catheter obstructions and hospitaliza- tions due to intermittent intracranial hypertension symptoms (symptomatic SVS) after both primary and secondary ASD implantation. Data in the study suggest that the high rate of ventricular catheter obstruction in pediatric shunt therapy is caused by hydrostatic suction induced by differential-pressure valve shunts during mobilization of the patient and that the development of a SVS can be traced back to this constant suction, which causes chronic CSF overdrainage and ventricular noncompliance. Recurrent ventricular catheter obstruction and SVS can be prevented by prophylactic supplementation of every shunt system with an ASD.
Conclusions. To inhibit chronic hydrostatic suction, to prevent overdrainage and proximal shunt obstruction, and to avoid SVS and thus improve the patient’s quality of life, the prophylactic implantation of an ASD in every pediatric hydrocephalus shunt is recommended.